Multi Jet Dry Type Plastic Cold Water Meter
|FOB Price:||US $7.6-8.3 / Piece|
|Min. Order:||1 Piece|
|Min. Order||FOB Price|
|1 Piece||US $7.6-8.3/ Piece|
|Production Capacity:||3, 000, 000/Year|
|Transport Package:||Export Standard|
|Payment Terms:||L/C, T/T|
- Model NO.: LXSC-15KS
- Medium Temperature: Cold Water
- Counter Instructions: Digital and Analogue
- Application: Civil
- Medium Pressure: General Water Meter
- Stucture: Traditional
- Type: Multi Jet Dry Type
- Temperature: 0 Degrees Celsius to 50 Degrees Celsius
- Dial Reading: 0.00002L to 9999 Cubic Meter
- Connecting Standard: Bsp & NPT
- Trademark: Aimei
- Origin: China
- Install: Horizontal
- Level Measurement: Class B/Class C
- Nominal Diameter: 15mm
- Principle: Speed of Water Meter
- Work Environment: Dry Water Meter
- Material: Plastic
- Size: 15mm
- Working Pressure: 16 Bar
- Pressure Loss: 63
- Length: 165mm/190mm
- Specification: Export Standard
- HS Code: 9028201000
It is a multi-jet super dry water meter for residential
application in sizes 15mm-50mm for cold meter.
Compact design with smaller meter body.
45° inclined register, hot wire tamper proof.
Magnetic drive mechanism with anti-magnetic protection.
External regulation device
Directive 2004/22/EC on measuring instruments;
the European Standard EN14154+A1+A2: Water Meter;
the International Organization of Legal Metrology OIML R-49: 2006(E)
Several lengths and connections available on request
Non return Valve
Reed switch option
Water temperature: ≤50ºC for cold water meter
Water pressure: ≤1MPa or 1.6MPa optional
(10bar or 16bar option)
Each meter has one set coupling optional: 2pcs coupling, 2pcs coupling nuts and 2pcs washers.
The meter should be installed in horizontal position with the direction of the flow as
indicated by the arrow cast in the meter body with the register face upwards.
Pipeline must be flushed before installation.
The meter should be constantly full of water during operation.
Maximum Permissible Error
In the lower zone from Qmin inclusive up to but excluding Qt is ±5%.
In the upper zone from Qt inclusive up to and including Qs is: cold water meter ±2%.